Required tools: Phillips screwdriver
The temperature and humidity sensor is installed in the radiation shield.
Loosen the two screws on the bottom of the radiation shield by three turns each. Now unscrew the lower part of the radiation shield using the large nut.
Pull out the sensor. The head of the sensor is plugged in. Pay attention to the markings (red dot, depressed tube/dent or, as in this case, a golden cross), as the sensor can also be turned by 180°. Pull off the old sensor.
Pull out the sensor. The head of the sensor is plugged in. Pay attention to the marking (red dot, depressed scrub tube/dent or as here golden cross), as the sensor can also be plugged in rotated by 180°. Pull off the old sensor.
Plug in the new sensor. Make sure that the labelling of the sensor is on the side with the marking of the plug and please only hold the sensor by the outer edges.
Check the function of the sensor at https://app.fruitweb.info or in the fruitweb app. New data is sent to fruitweb every minute so that the new sensor is recognised immediately.
Reinstall the sensor in the radiation shield.